An introduction to Triaxial Testing
Triaxial Cell Sand Sample Preparation
For some testing, system corrections may need to be applied that are not specific to a particular Instrument (e.g. Volume Change due to Cell and tube expansion in an Unsaturated Test). Alternatively, corrections may need to be applied to apparatus used in a Test that is not declared as an Instrument (e.g. frame deformation of an Oedometer).
Clisp Studio can create a secondary backup of the data captured as a test is running. The secondary backup is a file stored outside of Clisp Studio and can be used to recover data after a computer hardware failure.
PID settings are used in Clisp Studio in conjunction with a DSC2000 to fine tune the performance of a dynamic servo, both single (DSC2000 M) and dual axis (DSC2000 MM and HM). This support document is not designed to give a full accurate account of what PID’s are and how they are calculated.
When an installation and training package has been ordered with your VJ Tech system then the following instructions below should be undertaken before the engineer’s arrival to save time; which will ensure that the best use is made of the time available during your training sessions.
This blog explains how to Calibrate a Pressure Transducer for a DSC2000
Clisp Studio allows for Databases to be Attached and Detached whenever necessary.
The De-airing tank is a simple way of removing air from water and storing de-aired water in a isolated reservoir .
Before installing a new version of Clisp Studio, you will need to uninstall the current version.