PID settings are used in Clisp Studio in conjunction with a DSC2000 to fine tune the performance of a dynamic servo, both single (DSC2000 M) and dual axis (DSC2000 MM and HM). This support document is not designed to give a full accurate account of what PID’s are and how they are calculated.
When an installation and training package has been ordered with your VJ Tech system then the following instructions below should be undertaken before the engineer’s arrival to save time; which will ensure that the best use is made of the time available during your training sessions.
How VJ Tech automated equipment increased lab efficiency and client satisfaction for Professional Soils Laboratory (PSL, UK).
The De-airing tank is a simple way of removing air from water and storing de-aired water in a isolated reservoir .
Part One of our Triaxial Testing guide described the initial setup for a
Triaxial Test involving the System Setup, Sample Preparation and Test Creation. Part 2 described the setup required for the Saturation, Consolidation and
Shear stages. Triaxial Testing Part 3 will describe the Test execution, monitoring and results manipulation and assumes that all 5 setup
steps of the ‘Assistant’ have already been completed.
Part one of our Triaxial testing system setup guide explained the initial installation required that involves the System Setup, Sample Preparation and Test Creation. Now that Steps 1 and 2 (Hardware Setup and Specimen Details) of the wizard have been completed we will guide you through the necessary stage setup required for Saturation testing, Consolidation testing and Shear testing.
is used to examine the shear strength of different materials, including soil. It works alongside other testing methods such as Shear Testing and Consolidation Testing to ensure the limits of the material are understood and weaknesses are identified, optimising its performance when supporting a structural frame.